SINGCLEAN neutralization test for antibodies after blood vaccinationCode: 7695
Product detailed description
COVID-19 Neutralizing Antibody Test Kit CE Approved
The novel coronaviruses belong to the ? genus.COVID-19 is an acute respiratory infectious disease. People are generally susceptible.
Infection with COVID-19 initiates an immune response, which includes the production of antibodies in the blood. The secreted antibodies provide protection against future viral infections, because they will stay inthe circulatory system from months to years after infection, and will quickly and strongly bind to thepathogen to block cellular infiltration and replication. These antibodies are named neutralizing antibodies.
Sealed pouches each containing a test cassette, a desiccant
Lancets (for finger stick whole blood only)
Sterilizing tablet (for finger stick whole blood only)Buffer
lnstruction for use
Material Required but not provided:
1.Specimen collection containers
2.Centrifuge (for plasma only)
Storage and stability:
The kit can be stored at room temperature or refrigerated (4-30°C).The test device is stable through theexpiration date printed on the sealed pouch.The test device must remain in the sealed pouch until use.
DO NOT FREEZE.
Do not use beyond the expiration date.
After opening the sealed pouch, use the test as soon as possible within 60 minutes.
Test procedure :
Allow the test cassette, sample, buffer, and / or other materials to warm to room temperature (15-30 ° C) before testing. 1. Remove the test cartridge from the sealed foil pouch and use it as soon as possible. The best results will be achieved if the test is performed within 60 minutes. 2. Place the test equipment on a clean and level surface. For serum or plasma samples: Take a serum / plasma sample with the supplied dropper and then add 1 drop (approximately 20 ?l) of serum / plasma sample into the sample well (s). Then immediately add 2 drops of buffer to the sample well (S). Avoid formation air bubbles. For a whole blood sample: Using the supplied dropper, immediately transfer 1 drop (about 20 ?l) of whole blood to the sample well (S). Avoid formation air bubbles. 3. Wait until the colored line (C) appears. The result should be read after 10 minutes. Do not interpret the result after 20 minutes. 4. Use a disposable micropipette to draw blood. The blood level must reach above the narrowest part of the pipette. 5. Drop 2 drops of blood (or 1 drop of serum or plasma) into the smaller hole at the bottom of the test. 6. Add 2 drops of buffer reaction solution to the same hole. 7. The results can be read with the naked eye after 10-15 minutes (Note: Test results after more than 20 minutes are invalid!)
Interpretation of results NEGATIVE: If only lane C is present, the absence of any burgundy color in lane T indicates that none was detected in the sample neutralizing antibody COVID-19. The result is negative. POSITIVE: In addition to the presence of lane C, if lane T is shown, the test indicates the presence of the neutralizing antibody COVID-19 in the sample. The result is a positive neutralizing antibody COVID-19. INVALID: The control line does not appear. The most likely reason for a control line failure is insufficient sample volume or incorrect procedures. Check the procedure and repeat the test with a new test cartridge. If the problem persists, immediately stop using the test kit and contact your local distributor. Do not use after the expiry date. After opening the sealed sachet, use the test as soon as possible within 60 minutes.
Be the first who will post an article to this item!